|Title:||A simple methodology for characterization of germanium coaxial detectors by using Monte Carlo simulation and evolutionary algorithms||Authors:||González Guerra, Antonio
Rubiano, J. G.
González Guerra, Jonay
Arnedo, M. A.
Gil, J. M.
Bolivar, J. P.
|UNESCO Clasification:||221213 Radiación (electromagnética)
1206 Análisis numérico
Monte Carlo method
Characterization germanium detectors
|Issue Date:||2015||Publisher:||0265-931X||Project:||Evaluación Del Fondo Radiactivo Natural de Las Islas Canarias Orientales; Implicaciones Radiológicas Sobre la Población.||Journal:||Journal of Environmental Radioactivity||Abstract:||The determination in a sample of the activity concentration of a specific radionuclide by gamma spectrometry needs to know the full energy peak efficiency (FEPE) for the energy of interest. The difficulties related to the experimental calibration make it advisable to have alternative methods for FEPE determination, such as the simulation of the transport of photons in the crystal by the Monte Carlo method, which requires an accurate knowledge of the characteristics and geometry of the detector. The characterization process is mainly carried out by Canberra Industries Inc. using proprietary techniques and methodologies developed by that company. It is a costly procedure (due to shipping and to the cost of the process itself) and for some research laboratories an alternative in situ procedure can be very useful. The main goal of this paper is to find an alternative to this costly characterization process, by establishing a method for optimizing the parameters of characterizing the detector, through a computational procedure which could be reproduced at a standard research lab. This method consists in the determination of the detector geometric parameters by using Monte Carlo simulation in parallel with an optimization process, based on evolutionary algorithms, starting from a set of reference FEPEs determined experimentally or computationally. The proposed method has proven to be effective and simple to implement. It provides a set of characterization parameters which it has been successfully validated for different source-detector geometries, and also for a wide range of environmental samples and certified materials.||URI:||http://hdl.handle.net/10553/46045||ISSN:||0265-931X||DOI:||10.1016/j.jenvrad.2015.06.017||Source:||Journal Of Environmental Radioactivity[ISSN 0265-931X],v. 149, p. 8-18|
|Appears in Collections:||Artículos|
checked on Jul 25, 2021
WEB OF SCIENCETM
checked on Jul 25, 2021
checked on Jul 24, 2021
Items in accedaCRIS are protected by copyright, with all rights reserved, unless otherwise indicated.